How to fix disconnected mercury in sphygmomanometer
My sphygmomanometer suddenly has its mercury disconnected,meaning that there are like air gaps in it, which results in wrong readings, I think there is a way to fix it, but do not know how??
it's really important to me, and I almost daily need to use it. and do not want to buy another on.
Esta pergunta é pertinente?